Lower leg orthosis: Comfortable walker for ankle and foot injuries – in a lightweight shell construction
protect.CAT Walker short is lower leg brace. The product is exclusively to be used for the orthotic fitting of the foot and only on unbroken skin.
Safe and effective immobilisation
Three strap system ensures high stability and reduces anterior talar shift
Available as optional extra: heel wedge sets for Achilles’ tendon rupture
Open the two lower belts and the upper belt on the left-hand side. Fold the shin and the belts away to the right. Remove the inner shoe from the Walker.
Open the leg and foot part of the inner shoe by loosening the Velcro fasteners. Ask the patient to place his foot and lower leg into the inner shoe. Make sure that the air chambers are empty and that the valve is loosened anti-clockwise (deflate).
Close the inner shoe and make sure that it is firmly fixed to the patient's leg. Align the pump in such a way that it is sitting in the middle and so that later it fits into the space provided for the shin.
Now ask the patient to place their foot in the half shell of the Walker. The lower leg and foot should be positioned so that they fit exactly into the shell.
Check to see that the foot is properly set and positioned in the Walker. Make sure that the patient's toes are not pushing against the toe protector. If necessary, the toe protector's position can be adjusted or it can be removed completely.
Now place the loose shin onto the patient's shinbone and foot from the front. Make sure that the pump and ventilator fit into the spaces provided.
Close the upper belt.
Retighten the right-hand belt again as well in order to stabilise the system.
Then close all the other belts one after the other. The belts should fit tightly. Extra material can be cut away.
Ask the patient to walk around in the orthosis to make sure that the Walker is sitting properly.
To tighten the Walker's fit, the air cushion can be filled using the pump system. To this end, close the valve by turning it clockwise (inflate) and then pump air into the chamber. The padded pieces can be used as additional padding.
PP, PU, EVA, polyester
Wash by hand in mild, cold soapy water. Rinse thoroughly. Do not wring. Do not bleach. Air dry. Do not iron. Do not remove stains using benzene or chemical detergents. Remaining soap suds can cause skin irritation or can cause damage to the material.
- Postoperative: Post ligament reconstruction and/or refixation, for treatment of soft tissue injuries and tendon injuries
- Conservative: Ligament lesions 2nd and 3rd grade, tenosynovitis
- Stable foot and joint fractures: metatarsal fracture, talus fracture, malleolar fracture
Mon. to Thu. 08.00-17.00
- 02 9890 8696
- 02 9890 8439